Download for BioExplorer, BioEra, eBioo, Mindfulness Technology and Neurobit Recorder.
Download for BrainBay, OpenVIBE and applications created by users.
The channel profile "EEG 250sps" has been added. It simplifies configuration
of EEG measurement at 250 samples per second (2 fold higher, than for "EEG"
profile available so far).
For "EEG" and "EEG 250sps" channel profiles the "Common reference" field
is now set to "Reference 1" by default. This setting is suitable for
referential derivations (used most commonly).
Neurobit device settings, which have been already set by a superior
configuration means of an application, can now be locked (not available for
manual changes) in the settings window of the Neurobit Driver.
At the moment this feature is used by OpenVIBE software.
"Power mains frequency" setting (used e.g. by Neurobit Optima) is now read
from the system registry (and not from design or configuration file
maintained by an application). And it is set automatically, based on user's
geographic location (country) set in the system ("Control Panel"/"Regional and Language Options"/Location),
when the Neurobit Driver starts the first time, or when a user has changed geographic location
in the system since previous start of the driver.
Nevertheless, user can still change power frequency selection. It may be
necessary e.g. when traveling and location info in the system is not up-to-date, or in
Japan, where two power frequencies are in use.
This setting is now common for all applications using Neurobit Driver and
all users of a given computer.
Removed bug: in rare cases the application crashed while changing a channel
setting having dependencies (e.g. "Channel profile" or "Channel function"
influencing other channel parameters).
For interoperation with BioExplorer application (v. 1.6 or earlier), which does not
store Neurobit device settings along with a design, two buttons facilitating
device configuration have been added at the bottom of the window:
- "Load" settings from file,
- "Save" settings in file.
- Make sure that loaded configuration is suitable for
current BioExplorer design (e.g. that channels are configured to
measure EEG for a neurofeedback design).
- Only configuration file for current device model (selected in the Device
Manager of the application) can be loaded.
For Neurobit Optima devices, information about upper frequency limit of
temperature signal band has been updated according to changes in Optima
firmware ver. 1.1.
The "Sum disconnected" option is checked by default for the "Temperature"
Impedance bars on Test tab are now more accurate.
Background colors in device settings window have been corrected for eBioo application.
Status tab has been added in the device settings window during normal
measurement session. It enables to see detailed status (of each measurement
channel) if it is not shown directly in an application window.
Test tab is now hidden in the device settings window during normal
When "Device address" field is emptied manually,
the default address is automatically set.
Application can now read device serial number.
Channel parameter Common reference added for Optima devices.
It informs of external connection of "-" input of a given channel to
a common reference electrode. Its suitable selection is required for correct
results of electrode-skin impedance tests for such inputs. However, that
selection brings result for Optima devices with
firmware version 1.0.1
or newer (nevertheless, compatibility with Optima firmware 1.0 is preserved).
Help file for Device services has been included.
Download the latest driver version for your application using one of the links below.
If you have running an application supporting Neurobit equipment, exit it.
System administrator rights are required in this step. Unpack the downloaded archive to a suitable folder of your application, overwriting existing files.
Folders for driver files depending on an application (in relation to "Program Files" or "Program Files (x86)" folder):
|Neurobit Recorder ||Neurobit\Recorder